In this episode, Mike and Bob discuss a new tool for laser methane detection, which has been in the hazmat industry for a long time.
Thank to our sponsor, CBRNE Convergence World 2018. This show will be held in Orlando, Florida, USA. Click here for registration information and other fantastic information.
We’d also like to thank our sponsor, Flir, who is introducing the new identiFINDER R200. To find out more, click here.
Finally, thank you to our sponsor, First Line Technology. They’re debuting their new smartphone app called the Decon Field Guide. This app is available for both iPhone and Android.
We’re very excited to announce that our hazardous materials training manual, National Emergency Response Hazmat Drills: 50 Drills for Use with Hazardous Materials Personnel, is finally available on Amazon. This community-driven book features several volunteer contributors from the hazmat world who are located across the country. This book of drills is for you, and we hope you benefit from the hard work put in by everyone involved. Feel free to leave us a 5-star review on Amazon – let’s get this book on the bestseller list!
Complete Show Notes
3:45 What are Laser Methane Detectors?
- Like so many of our meters, laser methane detectors have their roots in industry
- Originally a piece of large, non-portable equipment consisting of a laser source and a detector
- Laser source and detector were placed in such a way that if a container (pipeline, cylinder, rail car, etc.) started to leak, the rising gas would go between the source and the detector and trigger an alarm
- As with most technology, this basic technology became smaller and smaller until we reached today’s topic – the portable laser methane detector
7:25 How Do Laser Methane Detectors Work?
- Each unit is equipped with 2-3 lasers
- Visible light laser
- Difference between 2 and 3 laser units depends on whether it has a visible light laser
- This type of laser acts as a visual cue to see the end point and is also a distance calculator
- Some detectors also use a screen similar to that of a thermal imaging camera – look through the screen and place a dot on the end target
- IR laser
- Methane has a very specific frequency and absorbs IR radiation, so one of these lasers will be set to frequency (most popular is around 3.3um)
- Reference laser
- Isn’t picked up by methane, so it passes through it unchanged
- Note that some units split one laser and use it as a reference, but the idea is the same – one for spotting/distance taking, one for reference, and one to hit/be absorbed by methane
- Keep in mind that the laser must always have something to bounce off of – if it doesn’t have anything to reflect off of, it can’t get the IR light back to the sensor
11:35 Example Scenario
- We shoot the laser across the room – it encounters a cloud of methane and adsorbs some of the light photons
- The rest of the laser moves through the cloud, hits a surface, and scatters
- Some of the light makes its way back to the meter, and the differences between the two are analyzed
- Detector gives us a reading of 500 ppm m.
- Keep in mind that these aren’t point source detectors – they’re only seeing the total distance the laser travels and the amount of light absorbed by the methane during travel
- We’re not measuring ppm in a specific spot like we would with a point source detector – instead, our reading represents the concentration through all the points in space the laser traveled (both high and low methane areas)
- This means that instead of just saying “ppm,” we have to clarify that it’s an averaged ppm based on distance traveled
19:10 Operational Uses of Laser Methane Detectors
- We are NOT using these meters to determine whether we’re in a flammable range
- We use laser methane detectors in three different ways:
- To determine if there’s methane present in a quick and safe way
- To determine if our actions produced a difference
- To triangulate where a gas leak is coming from (especially useful for underground leaks or leaks along pipelines)
- As you can see, there are many uses for these meters, but we’re never using them to determine if we’re in an LEL
26:25 Pros and Cons of Laser Methane Detectors
- Can cover large open areas, reducing the time spent on searching for leaks, as well as manpower
- Measurements can be made remotely, keeping team members out of harm’s way
- Calibration is minimal, and drift is non-existent
- Units respond almost instantaneously
- No cross-sensitivity – methane lasers only pick up methane
- Only picks up methane
- Corners really screw with readings, so make sure you’re not shining off of a corner
- Can’t use in open fields since there’s no spot for the laser to reflect back to you
- Unless there’s some kind of ceiling, you won’t get a reading outside above ground
- Can’t be used for gauging flammability
Have a question? Send an email to email@example.com or leave a message on our Haz Mat Guys comment hotline: 843-628-1484
- THMG014 – Metering Overview and Basic Sensors
- THMG015 – Advanced Metering
- THMG016 – Metering Techniques and Reading the Numbers